> the SWR shows really wrong values.. like 9:1 swr
> and as you key the radio, the SWR keeps changes. Sometimes, it shows
> correctly, other times the SWR keeps changing along with the keying
> anyone have any thoughts on this??
Yes, I have some thoughts, but they will not help you fix the problem.
Any SWR meter that directly displays SWR, without you having to manually
set the forward power to full scale first, has some kind of autoleveling
or calculation process going on. That process can be either analog or
digital. In either case, as long as the forward and reflected sense
voltages are sampled simulataneously and accurately, and the
"calculation process" is correct, the SWR displayed will also be
accurate. If on the other hand, if the forward and reflected sensing
occurs at slightly different times in a digital scheme, or with slightly
different time constants in an analog scheme, you will get inaccurate
results when your power is rapidly changing, as with SSB voice. Use a
constant level carrier (CW) and the reading will be accurate.
TenTec mailing list