My 259B needs some "minor" calibration for the "R" readings.
At a known good "pure" resistive load of 50 ohms, "R" reads 46.
I understand, per some TT threads that a precision test fixture is
required for "precise" calibration, but for now, I would like to at
least be in the ball park for say 8 ohms to 60 ohms at zero reactance.
Can anyone shed some light on the calibration procedures?
GET INTERNET ACCESS FROM JUNO!
Juno offers FREE or PREMIUM Internet access for less!
Join Juno today! For your FREE software, visit:
List Sponsored by AN Wireless: AN Wireless handles Rohn tower systems,
Trylon Titan towers, coax, hardline and more. Also check out our self
supporting towers up to 100 feet for under $1500!! http://www.anwireless.com
FAQ on WWW: http://www.contesting.com/FAQ/towertalk
Administrative requests: towertalk-REQUEST@contesting.com