I need some help in determining what the problem is,
in my Ham IV control box meter.
I pulled off the top cover to change the meter bulb and
since then the meter has been giving erroneous readings
Meter plate is South-center.
1. When power is off, I can't mechanically Zero the
needle. It keeps drifting either side of Zero, with very
small changes in the front panel screw. ( I'm using a
nonmagnetic plastic Bournes plastic tuning tool).
2. With power on, I can calibrate the meter ok.
3. The meter indicates OK from South (center), thru West
and up to North.
4. When rotating CCW from south, the meter will stay around
160 deg., then when the ant is stopped at 90 deg., the indicator
will catch up in a slow continuous sweep.
Note: I'm only turning a TH3jr and 3el 6 mtr yagi with this
rotor. I can also see the antenna from the shack, when
swinging it around.
5. From 90 deg, CCW to North, the needle will "lag" and slowly
creep towards North, but never go past about 15 deg.
6. Thinking the needle was binding in spots, I tapped the face of
meter (lightly) with my finger. The needle went crazy as if hit by
a static charge....
Next, I will try going thru the resistance and voltage checks to see if
I can get a handle on the problem, unless someone out there has
seen this condition before and has a solution.
I appreciate any and all help and ideas.
Thanks es 73,
GET INTERNET ACCESS FROM JUNO!
Juno offers FREE or PREMIUM Internet access for less!
Join Juno today! For your FREE software, visit:
This list is sponsored by the new eHam.net Store. When you buy products
like ICE filters, Array Solutions StackMatch, or M-Squared Antennas from
the eHam Store for the same price you pay direct, a portion of the sale
price goes to support this list. Check it out at http://www.eham.net.
FAQ on WWW: http://www.contesting.com/FAQ/towertalk
Administrative requests: towertalk-REQUEST@contesting.com