>
> Before I buy one, can you please advise whether I will be able to do
> this with an AEA CIA HF analyzer? It sure looks good on the
> manufacturers specifications. I had an Autek RF1 analyst before and I
> was quite dissapointed with its accuracy. What else do you use for
> tasks like this? Should I get the CIA or the VIA model?
>
>
> (Why do I want to measure so exactly? I want to determine the velocity
> factor of the wire used. The velocity factor is introduced by the
> insulation covering the wire. Therefore the wire elements are always a
> bit too long when cutting them to the dimensions taken from a computer
> simulation. This is not a problem with dipoles but with wire yagis and
> quads, where the parasitic elements must have exactly the proper
> resonant frequency.)
The length of feedline from the antenna to the measurement device
will affect the measured resonant frequency, so you also must be sure
the feedline is an exact multiple of 1/4 wl to determine EXACT
resonant frequency.
If it is not an exact multiple of 1/4wl, the line will shift the
reactance off center or even create a "false' indication of antenna
zero reactance. That's because the line will have some SWR, and the
SWR will cause reactance shift in the line. An odd 1/4 wl line will
invert the sign of the reactance, but will not alter it's absolute
value assuming the line is lossless.
The next problem you might have is line loss. Feedlines do not have
evenly distributed losses between conductor series resistance and
dielectric losses. The conductor loss dominates the system. The end
result of using a lossy feedline is it adds reactance to the system,
skewing the results.
You are trying to measure one of the most difficult things to measure
with nearperfect accuracy. You will probably learn more about how
difficult measurements like this actually are than you will about the
insulation. Be careful the measurement method you use and equipment
errors don't determine the results!
73, Tom W8JI
W8JI@contesting.com
