At 08:49 AM 8/12/05, Tom Rauch wrote:
>and read the warning at....
Tom, that has been corrected
in the online version. I don't recall it in
my '98 vintage manual either so your
warning about the 2002 revision sounds
From the Vf (p 21) and DTF (p 17)
procedures of the online manual:
3.) Tune the analyzer higher or lower in frequency until the Impedance
meter reads the very next impedance
minimum, and where reactance displayed on the LCD again crosses zero. A
non-zero minimum of a few
ohms is acceptable.
73, Bill W4ZV
See: http://www.mscomputer.com for "Self Supporting Towers", "Wireless Weather
Stations", and lot's more. Call Toll Free, 1-800-333-9041 with any questions
and ask for Sherman, W2FLA.
TowerTalk mailing list