The MFJ 259 has measures many functions.
It measures just about all transmission line equation parameters and the
alterations caused by loads - antennas.
Here's its parameter list and how to move to them.
In normal mode you have the starting parameters:
Impedance Frequency, R, X SWR
Press mode and the screen displays
For each additional mode tap of the mode switch it measures
and back to R,X,SWR,f.
Simultaneously press the mode and gate switches for a few seconds.
This enters you into the Advanced Mode.
Each tap of the mode switch offers:
Return Loss Coefficient
Distance to fault
Resonance mode tune
Z Magnitude and angle
One finds this by pressing the buttons and see what occurs or
Goggle MFJ 259 manual and you'll find sources of the manual. I believe MFJ.com
---- Kostas Stamatis <email@example.com> wrote:
> I have an analyzer mfj259. I am very happy with it but i think that i have
> not searched all of its features. For example, i recently learnt that i can
> use it to find coax loss. I know that maybe it was in manual but i had
> questions until i read here for the procedure and beside this i found that
> it is quite accurate (this is useful notice also). Beside this i found that
> many of us, who are lucky owners of 259, didn't know it also. So i would
> like to suggest here to make a list with uses of 259 and procedures also.
> Anoyone knows a relative site?
> 1) swr measure
> procedure: no need
> Accuracy: very good
> 2) Measure of coax loss
> procedure: connect the coax to 259. Keep the other side unshorted. Press the
> mode and measure.
> Accuracy: very good (2nd decimal point)
> Any more suggestions?
> Kostas SV1DPI
> TowerTalk mailing list
TowerTalk mailing list