Towertalk
[Top] [All Lists]

## Re: [TowerTalk] Why the inconsistent line impedance

 To: towertalk@contesting.com Re: [TowerTalk] Why the inconsistent line impedance K8RI Tue, 04 Sep 2012 15:36:07 -0400 mailto:towertalk@contesting.com>
 ```On 9/4/2012 11:57 AM, Jim Brown wrote: ``````On 9/3/2012 11:28 PM, K8RI wrote: ``````Can anyone tell me why the impedance is a sloping or irregular line where all the single band antennas have a very straight trace with the only blips and dips being connectors, the amp bypass, and the tuner bypass? `````` Several thoughts, Roger. First, TDR is usually done with a linear sweep, typically from 1 MHz to 500 MHz, or whatever the range of the analyzer. `````` ```I'm using the AIM 4170 and just learning how to use the various functions on it. Being a newcomer to the VNA I'm a complete novice learning how to use it, let alone how this specific VNA works. ``` ``````Because the sweep is linear, half of the energy in the sweep is in the top half of the sweep range, where small discontinuities in a line are more likely to be detected, but which do NOT relate very well to the impedance of a connected antenna in the operating range of the antenna. Second, the impedance we measure from one end includes reflections from the other end of the line. If you compare your red and green curves, their shape is nearly identical -- the same peaks and valleys, with the major difference between them being that the distance scale is more compressed. SO -- those peaks and dips are simply a reflection from the other end of the line. `````` ```I kinda though it was related to that, with each band having its own set of reflections and who knows what other odd reflections show up. Those reflections vectorially add. Even with a FFT "I think" the result would still be the equivalent of vectorially adding them. The last time I used FFTs was for photo enhancement in grad school and back then it took about 12 hours on the fastest 386 I could find.<:-)) ``` ```Using a mono-band antenna, without the multi-band antenna, Open, shorted, or with the 50 ohm standard the line is almost straight with tiny blips or rather dips showing up for connectors, the six-pack, and remote antenna switch while the amp and antenna tuner in the bypass positions show substantial peaks of about 73 ohms each and those do affect the tuning on 10 and six. ``` ```On both 40 and 6 the return loss is high as it should be. On 15 and 10 it's fairly low. I've paid scant attention to the WARC bands so far. ``` `````` Third, the TDR function in many analyzers (including mine) is an inverse FFT of the sweep data, and the shape of the trace will depend on the sweep rate, the excitation signal, how the data is filtered, and how it is windowed. My VNA, the DG8SAQ VNWA, allows me to change all of those parameters to maximize the sensitivity to defects. `````` ```There are a number of parameters I can change on the AIM 4170C, but as I said earlier, I'm a complete novice on the steeper part of the learning curve at present. ``` 73 Roger (K8RI) `````` 73, Jim K9YC _______________________________________________ _______________________________________________ TowerTalk mailing list TowerTalk@contesting.com http://lists.contesting.com/mailman/listinfo/towertalk `````` _______________________________________________ _______________________________________________ TowerTalk mailing list TowerTalk@contesting.com http://lists.contesting.com/mailman/listinfo/towertalk ```
 Current Thread [TowerTalk] Why the inconsistent line impedance, K8RI Re: [TowerTalk] Why the inconsistent line impedance, Jim Brown Re: [TowerTalk] Why the inconsistent line impedance, K8RI <=