TenTec
[Top] [All Lists]

## [TenTec] Re: watt meter calibration

 To: [TenTec] Re: watt meter calibration w5yr@worldnet.att.net (George T. Baker) Fri, 17 Dec 1999 19:50:44 -0600
 ```Regardless of the diode rectifier used, you have to be careful about the value of the resistor used across the filter cap and the input impedance of the DVM. If the DVM loads the circuit too much, you get less than the peak voltage and calculate less than the true power value. Most probes are designed to take into account the DVM loading such that the output voltage is about 0.7 of the peak to give an approximate RMS value. The Handbook probes, for example, were designed that way so that the resulting power calculation was simply E^2/50 where E was the voltage read by the DVM. If I remember right, the load resistor was chosen for an 11 megohm input resistance for the VTVM that used to be the standard. Of course, if you can confirm that you are actually seeing the peak voltage, then the calculation is (Ep^2)/100 where Ep is the true peak voltage across the 50-ohm resistive load. 72/73, George Fairview, TX 30 mi NE Dallas in Collin county Amateur Radio W5YR, in the 54th year and it just keeps getting better! R/C since 1964 - AMA 98452 RVing since 1972 Roy Koeppe wrote: > > About: > > "I have an RF probe made per ARRL hand book with a 1N34A diode which is > good to about 20 volts. I use it for alignments and such. With 20 > volts > RF into a 50 ohm load I have 8 watts carrier out. This is pretty close > with what I have on the watt meters as well but too low to calibrate > from. Can I put 10 of these diodes in series and run the RF volts to 200 > volts and expect to get 80 watts on the watt meter? Is this an accurate > way to calibrate an RF watt meter or is there a better way?" > > I think there is an improved (accuracy) way. You can use a 6AL5 tube > with its heater tied to its cathode, and its two plates connected in > parallel. This configuration allows for a full legal power rating. Use > two caps in the filter, one for RF and the other bigger one for a > holding factor. Measure the DC voltage output with a VOM or VTVM, and > convert the reading to RF watts using the standard peak voltage formula. > Mine works great from QRP to QRO. Accuracy depends on your meter, > typically 2-percent, etc. > > 73, Roy K6XK Iowa Outback -- FAQ on WWW: http://www.contesting.com/tentecfaq.htm Submissions: tentec@contesting.com Administrative requests: tentec-REQUEST@contesting.com Problems: owner-tentec@contesting.com Search: http://www.contesting.com/km9p/search.htm ```
 Current Thread [TenTec] Re: watt meter calibration, Roy Koeppe [TenTec] Re: watt meter calibration, George T. Baker <=