On Wed, 31 Jan 1996, Rein Couperus wrote:
> I am having EXACTLY the same problems with TR-LOG.. it is definitely
> a defect in the OMNI firmware. Interestingly, on 4800 bd the keyer lockup
> does not happen, only on 1200 and 2400 bd. BTW, the lockup not only happens
> to the internal keyer, also the external keying line is DEAD AS A DODO.
> The only remedy is switching the rig on and off.
> Another interesting feature of the OMNI VI: when I try the computer interface
> with 1200 or 2400 Bd, the tuning rate goes down to 85 Hz per rev. This can
> be changed by a MASTER RESET. Also this effect is not present at 4800 Bd..
> Hope this helps,
> -Rein, PA0RCT
> http://www.iaehv.nl/users/reinc (link to KM9P included...)
I am using CT both version 8 and 9, I also have NA version 9 and I have
not run into this problem, yet. I have Cde turned on for both my every
day logging program and for CT, so far it has not bothered NA. The
display has not gone haywire yet. There is a possibility that I might
have older firmware, my rig was purchased in November 1992, and serviced
in November 1994. I have used it for our multi-multi operations at N2RM,
and have had no problems. I run my interface at 9600 baud usin a 3
conductor cable between computer and rig.
Warren - N2BCC