[TenTec] Argonaut VI Lost RX/TX on 160/80 Meters
rohre at arlut.utexas.edu
Thu Jul 2 15:26:24 EDT 2015
Failures where they do not do closer than board isolation are concerning
to us hams. We all want to know what and why the failure?
Sounds like you were doing most things to prevent lightning damage.
However, have you had some windy days just before the failure happened?
You might have been the victim of wind static on a wire or even a beam
Adding the surge protection and the bleed resistor are good ways to
discharge any wind static build up before it reaches critical voltage.
I had not thought about the new generation of rigs having such small
components that diagnosis might not be possible without special tools.
It leads me to conclude, to fit all future antennas with static bleed
For Field Days when our club ran multi-wavelength loops,
We successfully used 150 K ohm 2 watt or 1 watt resistors from each side
of the balanced feeder to a ground lead to a metal tent stake driven
completely into the soil beside the operating position.
The year prior to the adoption of that safety aid, we had wind static
from a storm 10 miles away arc through the plastic on a wire nut to a
grounded point. The plastic wire nut insulator was rated at 600 volts.
More information about the TenTec