[TowerTalk] Measuring Common Mode Choke with Anritsu VNA
wb4rtp at mindspring.com
Tue Jan 2 23:34:54 EST 2018
On 1/2/2018 10:00 AM, Jim Brown <jim at audiosystemsgroup.com>:
> Somehow I failed to clearly make the point that measurement errors
> associated with measuring high Z chokes with S11 is NOT a relatively
> small error, like 10%. It can be a VERY LARGE error, like 50-70% in
> resonant frequency. Which FAILS to allow assessment of suitability for
> use in a given frequency range.
I agree the error can be very large, but whether that error is
significant depends on the application. The Keysight charts clearly
show that using a professional-grade VNA, the S11 measurement can be
accurate up to 2,000 Ohms impedance, and that can be enough for many
common-mode choke application. If I only need 1,000 Ohms impedance for
my application I don't care if the actual impedance is 1,000, 10,000 or
100,000 Ohms. At the very least, the S11 Smith chart test is a good
quick sanity check that a choke is basically working. If I need to
prove a choke has a lot more than 1,000 Ohms impedance, then I agree
that some other technique is needed, and the Keysight charts discuss
those other methods.
Again, I thank you for bringing this to my attention, prompting me to
search for impedance measuring techniques and leading me to these
This email has been checked for viruses by Avast antivirus software.
More information about the TowerTalk