Probably don't want to start messing with the slug calibraton.
You could drive a current sink (op-amp + JFET + 420 ohm R) with a 1400 ohm
input R to match the type 43 slug.
Something like this:
Note that operating at such low currents that the NPN booster device is not
A very low offset op-amp is required. Single supply may be possible.
73 & Good morning,
--- On Tue, 9/6/11, email@example.com <firstname.lastname@example.org> wrote:
> From: email@example.com <firstname.lastname@example.org>
> Subject: [Amps] bird meters
> To: email@example.com
> Date: Tuesday, September 6, 2011, 8:25 PM
> I have a couple of BIRD 100ua meters
> 2150-090 es 091
> if I use these on my thruline 4522-002-8 50 ohm with 43
> slugs wot
> error will I see?? as these slugs are for 30ua..meters!!
> or can I recal the slugs es if so "how"
> tu 73 vk3pa
> Amps mailing list
Amps mailing list