[TenTec] S-Meter Calibration in 3.033xa
cw_de_n5op at sbcglobal.net
Sun Apr 7 19:02:59 EDT 2013
And I must also be more tired than I thought since I'm really concerned with the S-meter, not the A-meter...
"People that make music together cannot be enemies, at least as long as the music lasts." -- Paul Hindemith
On Apr 7, 2013, at 17:45, Kim Elmore <cw_de_n5op at sbcglobal.net> wrote:
> I must have missed the memo because I just downloaded and installed 3.033xa for the O II. I calibrate0d my S-meter as per the instructions and.. Wow. My values are quite different for the default values. For example, in my case I need 107 counts for S-9+50 dB while the default is 221. S-9 is displayed for me with 55 counts while the default is 89. All of mine are much closer together than the default values.
> Since I obviously missed this announcement, I'm curious: is this a function of the meter movement or something else? I interpret this making my S meter much more sensitive compared to the default values. Does this calibration affect the AGC itself, or only the displayed meter value?
> Kim N5OP
> TenTec mailing list
> TenTec at contesting.com
More information about the TenTec