[TowerTalk] 259 Grid dip meter adapter
Sun, 17 Sep 2000 19:55:31 -0400
> I also have found the 259 GD coil to be vy insensitive. I have used both
> the ac powered and battery powered Millen GDO's to cut 1/4 wave stubs.
> They work fairly well but there is a better way reccomended to me by K5GW
> of Tx Towers. It is used by VHFers.I understand he used it to make the
> stubs in his 2m moonbounce array. I believe it is 48 each 10 element
The 259 GD adapter is an offshoot of a QST article (I think that
article won some award).
Unfortunately, it does not work well.
It can't work well because the coil in the adapter is not part of a hi-
Q resonant circuit. If the external system is high-Q, the adapter will
work OK. If the external circuit is low Q, it will not. Stubs are
generally low to moderate Q devices.
The 259 works better checking stubs by watching the "Impedance"
meter. The 259B reads stubs directly.
IMO, the GD adapter isn't very useful with the 259.
73, Tom W8JI
FAQ on WWW: http://www.contesting.com/FAQ/towertalk
Administrative requests: towertalk-REQUEST@contesting.com